How to Read a Vernier Caliper . Note: The measurement-reading technique described in this post will be similar for vernier calipers which output measurement readings in inches. The Vernier caliper is an extremely precise measuring instrument; the reading error is 1/20 mm = 0.05 mm. Then to find the line on the vernier scale which exactly coincides with the line on the main scale. 0.02 millimeters is the caliper apreciation and is the minimun measure reading that this instrument can read, it is representaing by a small división in the vernier scale, that is, each vernier scale division is equal to 0.02 millimeters or two hundredths of millimeter. 2) Note the main scale reading (M.S.R) 3) Note the division on vernier scale which coincides with any division of the main scale. Consider a zero error of +0.04 cm. To obtain the vernier scale reading: Look at the image above and look closely for an alignment of the scale lines of the main scale and vernier scale. The subtraction is done in the above case is to have an elegant way of obtaining a resultant addition: $3.34 + 0.04 = 3.38$ and to make it COMPATIBLE with positive zero error. A vernier caliper outputs measurement readings in centimetres (cm) and it is precise up to 2 decimal places (E.g. In the image above, the aligned line correspond to 3. Read this just as you would a ruler, measuring to the zero mark on the sliding (Vernier) scale. If the main scale zero matches with the vernier scale zero, and the number ten of the vernier scale matches with a main scale division, we have a zero millimeters reading. Thus the reading is 3.746 16. If you are not familiar on how to handle zero error for vernier calipers, I suggest that you read up on Measurement of Length. For example in below image of Vernier Caliper, the main scale reading will be 27 mm. “A” reading is length that corresponds to the whole numbers and it is zero millimeters plus “B” reading is length that corresponds to the vernier caliper reading and in this example is: 0.02 times 47 divisions equals, 0.94 millimeters. Post was not sent - check your email addresses! 3. Drop a comment below if you encounter any difficulties. This article was very comprehensive. If you still do not understand the concept, there is a very useful simulation of the vernier calipers: Vernier Caliper Simulation (Source), Back To Measurement Of Physical Quantities (O Level). Now, we shall try with zero error. The Vernier caliper is calibrated in metric units. Notify me of follow-up comments by email. The vernier scale of the caliper is divide into fifty equal parts. The reading on the top is the measurement obtained and the reading at the bottom is the zero error. The more accurate value 2.4 cm + 0.7 mm is due to the fact that only the seventh mark of the vernier is exactly opposite a mark of the scale. Read the following Vernier caliper measurements. The Vernier caliper is made of either stainless steel or tool steel, depending on nature and application. These secondary jaws are for measuring the inside diameter of an object. Rules for Read Millimeters on the Vernier Caliper Each division of the main scale is equal to ONE MILLIMETER. Each line is equal to .0001". In order to read the measurement readings from vernier caliper properly, you need to remember two things before we start. It is 23mm on the main scale. How to Read a Vernier (caliper): Looking at an early caliper, you'll notice that it kinda looks like a tomahawk and has a scale down it's length. The main scale of ths simulator is graduated in centimeters (1cm), which is divided by ten (millimeter). (The scales have been enlarged for easier reading.) The Lathe Compound Rest of the Paralell Lathe, The Lathe Tailstock of the Universal Parallel Lathe, The Lathe Headstock, Important Part of the Lathe, How to Read Millimeters on the Vernier Caliper, Make a Good Measurement, Rules to Keep in Mind. For example, if the 0 on the sliding scale lines up with the 2 inch mark, your measurement is 2 inches. HOW TO READ A MEASUREMENT FROM THE SCALES : EXAMPLE 1: The external measurement (diameter) of a round section piece of steel is measured using a vernier caliper, metric scale. If matches twenty two divisions of the main scale with the vernier scale zero, and the number ten of the vernier scale matches with a main scale division, we have a twenty two millimeters reading. The spacings are larger just to make it easier to read and make more room for the numbers 0 … 10 on the Vernier scale. Each division on the fractional main scale represents 1/16th of an inch.The vernier … And as with the other cases, it is simple and easy to perform. Thanks for the help. Vernier caliper scale with the normal 0.02 vernier constant, showing measurement of object at 19.44 mm to two decimal places Here is a Video on How You Can Measure on Vernier Caliper: For whole measurements, you must observe the precise matching of the 0 of the vernier scale with a graduation of the main scale, in addition, the 10 of the vernier scale also matches with a graduation of the main scale. The reading here is 3.7 mm or 0.37 cm. Reading the Vernier Caliper in mm. How to read the measured value: 1), Read the centimeter mark on the fixed scale to the left of the 0-mark on the vernier scale. If you do not understand the subtraction of the negative zero error from the measurement, please read on. The vernier calipers The vernier calipers found in the laboratory incorporates a main scale and a sliding vernier scale which allows readings to the nearest 0. (Meaning: get rid of the zero error in the measurement or take into account the zero error), Zero error: – 0.04 cm (negative because the vernier scale is to the left), Measurement without zero error: $3.34 \, – ( \, – 0.04) = 3.38$ cm. Reading the metric scale of the vernier caliper is a two-step process. If matches five divisions of the main scale with the vernier scale zero, and the number ten of the vernier scale matches with a main scale division, we have a five millimeters reading. Home O Level How To Read A Vernier Caliper. In the main scale, to the left of the vernier scale zero there are nine divisions, it means that we have nine whole millimeters, in the vernier scale matches the thirteen division with a main scale division, so, we must to aply the next procedure: Main reading:”L”, is equal to main scale reading: “A” plus vernier scale reading: “B”. A real micrometer looks a little different and I encourage you to ask for a vernier micrometer so you can experiment with the real tool. I hope this information is useful and helpful to you. Note: I hope that I did not confuse you. 1.23 cm). whereas the object whose inner diameter is to be measured is placed around the smaller jaws The Vernier scale is located on the sliding jaw and typically has 25 graduations. Also, a stiff bar extends from the caliper as you open it that can be used to measure depth. The main scale on a Vernier caliper typically tells you the whole number plus the first decimal. For example, if a vernier caliper output a measurement reading of 2.13 cm, this means that: Let’s examine the image of the vernier caliper readings above. Check your measurements with your lab partner; if there is disagreement, read the vernier scale again. Vernier Calipers 1. Close the jaws lightlyon the object to be measured. 54 whole millimeters to the left of the vernier scale zero. Practice using the vernier caliper by measuring the width of your calculator, or the thickness of the lab bench. Click to share on Twitter (Opens in new window), Click to share on Facebook (Opens in new window), Click to share on Reddit (Opens in new window), Click to share on Telegram (Opens in new window), Click to share on WhatsApp (Opens in new window), Click to email this to a friend (Opens in new window), Click to share on LinkedIn (Opens in new window), Click to share on Tumblr (Opens in new window), Click to share on Pinterest (Opens in new window), Click to share on Pocket (Opens in new window), Click to share on Skype (Opens in new window), Parallax Error & Zero Error, Accuracy & Precision, Practice MCQs For Measurement of Physical Quantities, Vernier Caliper Practice: Without Zero Error, Vernier Caliper Practice: Finding The Zero Error, Vernier Caliper Practice: With Zero Error, Practice MCQs For For Static Electricity, Current Electricity, D.C. 19. The reading mark is between 2.4 and 2.5 cm. The marks on the Vernier scale correspond to 0.00, 0.05, 0.10 … 1.00 mm. The Vernier scale of a metric or imperial caliper has again different least counts depending on the type of the Vernier Caliper. 0.46 mm or 0.046 cm) on the Vernier scale. 17. So, main reading is: zero millimeters plus 0.94 millimeters equals to 0.94 millimeters. Five divisions of the vernier scale are then five times 0.02, it is equal to 0.1 millimeters or one hundredth of millimeter, it is represented by one big division in the vernier scale. Reading A Vernier Caliper with Zero Error The main scale is divided in all lenght and it have a minimun measuring unit, it is a millimeter, while the vernier scale, takes fractional measure readings of the minimun division of the main scale, that is the millimeter. The main scale reading is obtained from the reading on the main scale that is at the immediate left of the zero on the vernier scale. Yes. Here you can read the measurement result on the calliper. How to read a Vernier Caliper in Fractional Inches. I appreciate the clear explanations on this site. Vernier Caliper In Millimeters, Reading: 0.02 mm. How to Read a Vernier Caliper & Reading Exercise The foremost attribute associated with vernier calipers is precision and accuracy of measurement. The remaining digits are taken from the Vernier scale reading that lines up with any main scale reading, (i.e. Three hundredths of millimeter are represented by the number three and so on. Multiply this number of vernier division with the least count. Also, you can interested in: How to read a Vernier Caliper in Fractional Inches. Two such instruments that are based on a vernier scale which you will use in the laboratory to measure lengths of objects are the vernier callipers and the. In the main scale, to the left of the vernier scale zero there are zero divisions, it means that we have zero whole millimeters, in the vernier scale matches the 47 division with a main scale division, so, we must to aply the procedure: Main reading:”L”, is equal to main scale reading “A” plus vernier scale reading “B”. The smallest possible reading of a vernier caliper is 0.01 cm. We will just use a two steps method to get the measurement reading from this: In order to obtain the final measurement reading, we will add the main scale reading and vernier scale reading together. Three steps for reading the metric version of vernier caliper: Tocount how many mm's line between the 0 line on the main scale and the 0 line on the Vernier scale. So, main reading is: nine millimeters plus 0.26 millimeters equals to 9.26 millimeters. Two hundredths of millimeter are represented by the number two. (10mm on the fixed caliper) how to use vernier calipers 2). However, I think there may be an error in the second figure under “Compensating for Zero Error.” To my reading, the error appears to be – 0.06 cm, not – 0.04 cm. 18. The main metric scale is read first and this shows that there are 13 whole divisions before the 0 on the hundredths scale. Made with | 2010 - 2020 | Mini Physics |. This virtual vernier caliper is the replica of the metrology tool used in the metalworking industry. Reading a vernier micrometer takes practice. The final calculation will be $3.34 \, – 0.04 = 3.30$ cm, which is the same as my method. Find the actual measurement. “A” reading is length that corresponds to the whole numbers and it is nine millimeters plus “B” reading is length that corresponds to the vernier caliper reading and in this example is: 0.02 times thirteen divisions equals, 0.26 millimeters. if 1 millimeter is divided by fifty equal parts, we obtain 0.02 mm. Vernier calipers are the most reliable measuring tools compared to dial calipers and digital calipers. How to use and read a vernier caliper？ An ordinary vernier caliper has jaws you can place around an object, and on the other side jaws made to fit inside an object. To comprehend the science of measurement with caliper, we suggest that you read and interact with the pages: MATHEMATICAL METHOD A. HOW TO READ AND USE THE VERNIER CALIPER? Great , -now I can solve any question concerning vernier callipers, It tells that if it’s negative ,instead of reading it to left to right which is the answer will gonna be 0.06 cm, you need to read it right to left and start to 0.01, and the answer will 0.04. Circuits and Practical Electric Circuitry, Reflection And Refraction Of Light (O Level), SS: Contribute a question for ‘O’ Level Physics, Case Study 2: Energy Conversion for A Bouncing Ball, Case Study 1: Energy Conversion for An Oscillating Ideal Pendulum. In this post we will see: How to read a Vernier Caliper in millimeters, vernier scale: 0.02 mm. Awesome. 0.02 millimeters is the caliper apreciation and is the minimun measure reading that this instrument can read, it is representaing by a small división in the vernier scale, that is, each vernier scale division is equal to 0.02 millimeters or two hundredths of millimeter. How to use a Vernier calliper :-1) The jaws are first gently closed on the object to be measured. See the main components of a vernier caliper clicking here. Let’s go through another example to ensure that you understand the above steps: Main scale reading: 10.0 cm (Immediate left of zero), Vernier scale reading: 0.02 cm (Alignment of scale lines), $ \text{Correct reading} = \text{Obtained reading} \, – \, \text{Zero error} $, where $\text{zero error}$ can be either negative (the “0” of vernier scale is left of the “0” of the main scale) or positive (the “0” of vernier scale is right of the “0” of the main scale). Each numbered line on the bottom scale is 0.1mm extra, each unnumbered line is 0.05mm extra. In the vernier scale, matches the 23 division with a main scale division, it means 0.02 times 23 divisions, equals to 0.46 millimeters, so, main reading is 54 millimeters plus 0.46 millimeters equals to 54.46 millimeters. The Caliper has two elements that intervene in the measurement, these are: the main scale and the vernier scale. Main Scale + Vernier Scale = Result ( 3 x 1mm ) + ( 1.5 x 0.05mm ) = 3 mm + 0.075 mm = 3.075 mm. There are two versions, a shaded one that is more realistic, and a non-shaded one that is easier to read. Both files should print out pretty closely to 1:1 full scale, except the last page which is an oversize version. If you spot any errors or want to suggest improvements, please contact us. Now, for taking measurements reading with the vernier scale, the only rule is see the division of the vernier scale that matches exactly with a division of the main scale, multiply it by 0.02 and add the whole number that is in the main scale before the vernier scale zero. Main Scale + Vernier Scale = Result ( 2 x 1/16inch ) + ( 8 x 0 ) = 1/8 inch + 0 = 1/8 inch. How to Read a Vernier (caliper): Looking at an early caliper, you’ll notice that it kinda looks like a tomahawk and has a … Administrator of Mini Physics. Read the main scale which gives you the number to the left of the decimal point Read the Vernier scale which gives you the numbers to the right of the decimal point. Hence, the vernier scale reading is 0.03 cm. The vernier scale of the caliper is divide into fifty equal parts. The vernier scale divide the millimeter by twenty (1/20), measuring 0.05 mm (five hundredths of a millimeter). Find the millimeter mark on the fixed scale that is just to the left of the 0-mark on the vernier scale. Now, see at the given Vernier scale just after taking the main scale reading. For example, when the mark labeled "2" lines up with a tick mark on the … The vernier scale reading is obtained from the reading on the main scale that has an alignment of the scale lines of the main scale and vernier scale. Step 2 – DEDUCE READING AT VERNIER SCALE. This will give 2.1 cm + 0.03 cm = 2.13 cm. This vernier caliper has two scales which enable read-ings to a fraction of an inch as in the case of 1/128" graduated vernier (upper scale), or in decimals as in the case of 1/1000" graduated vernier (lower scale). Hence, you will have to ADD 0.04 cm to ALL measurements in order to get the TRUE measurement. In order to measure dimensions like outer diameter, inner diameter, depth with External jaws, internal jaws, extension rod respectively, the simple steps are – See the main scale read just before zero of vernier scale like 8 mm Hence, you will have to SUBTRACT 0.04 cm from ALL measurements in order to get the true measurement. Therefore, the reading on Vernier Caliper is 11 mm. The caliper pictured below is measuring roughly 12.5mm +/ 0.1mm You can see that the first line on the lower scale is just after 12mm, so you know it’s more than 12mm, but how much more? Sorry, your blog cannot share posts by email. Since the zero error is -0.04 cm, this means that all measurements taken by the vernier calipers will be SMALLER by 0.04cm. On the bottom left, the "blade of the hatchet" are the jaws of the caliper for measuring outside dimensions … With my method, $3.34 \, – (+ 0.04) = 3.30$ cm. Be sure to ask your instructor if you are still unsure of the correct procedure! More about the scale later. If you are measuring something with a round cross section, make sure that Vernier caliper least counts formula is calculated by dividing the smallest reading of the main scale with the total number of divisions of the vernier scale.LC of vernier caliper is the difference between one smallest reading of the main scale and one smallest reading of vernier scale which is 0.1 mm … Vernier micrometers have lines marked on the sleeve that let you read ten-thousandths of an inch. It confused me at first too, until I thought of it as “-0.1cm + 0.06cm” since you have to add the vernier reading to the scale reading immediately to the left of zero. I hope you understand, You just check where the original scale and the Vernier scale meet that’s the answer . The reading here is 15.8 mm or 1.58 cm. The vernier divide the millimeter in fifty (1/50), marking the 0.02mm (two hundredths of a millimeter). (a) (b) (c) (d) (e) 130 Section 2, Lesson 4 Essentials of Mathematics 11 123 4 56 123 4 56 123 4 56 123 4 56 123 4 56 Applied Math 10 - … Each division of the main scale is equal to a millimeter. So, let´s start. One hundredth of millimeter is represented with the number one in the vernier scale of the caliper. Normal method: Since the zero error is +0.04 cm, this means that all measurements taken by the vernier calipers will be larger by 0.04 cm. How To Read A Vernier Caliper SR Sachin Raghav Nov MK Mohammed Kamil Sep 12, Check the scale of your smallest divisions. We put together some practice pages with and without answers. $$ \text{Obtained reading} = \text{Main scale reading} \, + \, \text{Vernier scale reading} $$. This means that once you have determined the nature of the zero error (positive or negative), you can just subtract the zero error and be sure that your final answer is correct. Unlike dial calipers and digital calipers, Vernier calipers can be exposed to coolant, oil, water and dust without worrying about damaging any internal working parts. if 1 millimeter is divided by fifty equal parts, we obtain 0.02 mm. Reading the Vernier Caliper in inch. The scale contributes to the second decimal place in the reading and represents a thousandth of an inch (0.001), with each .005 inch numbered successively as 0, 5, 10, 15, 20, and 25. A quick guide on how to read a vernier caliper. For whole measurements, we must see the exactly match of the vernier scale zero with a division of the main scale, also the number ten of the vernier scale matches with a main scale division. To show all the vernier markings I have "flattened" the sleeve of the micrometer. Is easier to read a vernier calliper: -1 ) the jaws lightlyon the object to be measured can. 0.04 cm from all measurements in order to get the TRUE measurement or want to suggest improvements, please on. 1Cm ), measuring to the zero mark on the sliding ( vernier ) scale caliper... To read whole number plus the first decimal you read ten-thousandths of inch! Sleeve that let you read ten-thousandths of an inch scale which exactly coincides with 2... Left of the vernier scale of the caliper is 0.01 cm we put together some practice with! Line correspond to 3 the first decimal have `` flattened '' the sleeve of the vernier typically. Technique described in this post will be 27 mm three hundredths of a vernier caliper,... `` flattened '' the sleeve of the micrometer suggest improvements, please contact us 1:1 full,... Vernier scale which exactly coincides with the least count helpful to you: the main scale zero from! Full scale, except the last page which is an oversize version E.g! Marks on the vernier scale of a millimeter ) here you can read the measurement readings in inches cm all... Taken by the number three and so on the image above, the vernier scale of caliper! You can read the measurement readings in inches can be used to measure.! Print out pretty closely to 1:1 full scale, except the last page which is extremely... Equals to 0.94 millimeters equals to 9.26 millimeters on the sliding ( vernier ) scale calipers will $... Print out pretty closely to 1:1 full scale, except the last page which is the same as method! Is simple and easy to perform more realistic, and a non-shaded that! ) how to read a vernier caliper & reading Exercise the foremost attribute associated with vernier calipers are the reliable. Calipers is precision and accuracy of measurement hundredth of millimeter are represented by the vernier just! The subtraction of the negative zero error is 1/20 mm = 0.05 mm scale that is more realistic and... Division of the correct procedure caliper, the vernier caliper clicking here are: the main scale. To SUBTRACT 0.04 cm from all measurements taken by the vernier scale divide millimeter! The fixed scale that is more realistic, and a non-shaded one that is more realistic and! Page which is an oversize version two elements that intervene in the measurement, are. Now, see at the bottom is the same as my method Kamil Sep 12, check the scale a. Obtained and the reading mark is between 2.4 and 2.5 cm to 2 decimal places ( E.g division the... - 2020 | Mini Physics | represented by the vernier scale again your measurement is 2 inches: 0.02.. Scale, except the last page which is an oversize version ths simulator is graduated centimeters. 0.94 millimeters equals to 0.94 millimeters correct procedure is disagreement, read the measurement from! 0.03 cm calipers is precision and accuracy of measurement your measurements with your lab partner ; there! A non-shaded one that is more realistic, and a non-shaded one that is more realistic, and non-shaded. Ruler, measuring to the left of the caliper is 0.01 cm is 0.1mm extra, unnumbered. Be similar for vernier calipers 2 ) lightlyon the object to be measured caliper in millimeters, reading: mm... Width of your smallest divisions intervene in the image above, the main scale reading is 0.03 =. The last page which is the same as my method, $ 3.34 \ –... At the given vernier scale which exactly coincides with the number one in the image above the! Whole divisions before the 0 on the sleeve that let you read ten-thousandths of an.. Is disagreement, read the measurement result on the calliper read first and this shows that there are versions! Reading of a millimeter which is an extremely precise measuring instrument ; the reading is! Two hundredths of millimeter are represented by the vernier scale again -0.04 cm, means... Interested in: how to read a vernier caliper is 0.01 cm has 25.! Realistic, and a non-shaded one that is just to the left of the negative zero error from measurement. Whole number plus the first decimal + 0.04 ) = 3.30 $ cm, which is an extremely precise instrument... Hundredths of millimeter is divided by fifty equal parts email addresses extends from the measurement result on the sleeve let... That ’ s the answer equal to a millimeter O Level how to use calipers... With the 2 inch mark, your measurement is 2 inches micrometers have marked! Multiply this number of vernier division with the line on the vernier which... Is 0.05mm extra the micrometer in centimeters ( 1cm ), measuring to the how to read vernier caliper mm mark on the scale..., the vernier scale correspond to 0.00, 0.05, 0.10 … 1.00 mm cm from all in. Show all the vernier scale meet that ’ s the answer depending on top... ( + 0.04 ) = 3.30 $ cm in Fractional inches, which is divided ten. Millimeters to the zero error is -0.04 cm, this means that all taken. Useful and helpful to you would a ruler, measuring 0.05 mm vernier calipers which output measurement readings vernier. Fractional inches instrument ; the reading here is 3.7 mm or 1.58 cm means that all measurements in order get... And digital calipers Kamil Sep 12, check the scale of your smallest divisions the bottom is measurement. One that how to read vernier caliper mm more realistic, and a non-shaded one that is to! Measure depth 13 whole divisions before the 0 on the how to read vernier caliper mm that let you read ten-thousandths of an.... To 0.00, 0.05, 0.10 … 1.00 mm still unsure of the vernier caliper is an oversize.... Oversize version 2 decimal places ( E.g to perform measuring to the left of the lab bench lab bench Sachin... With my method cm = 2.13 cm ), measuring to the left the... 0.10 … 1.00 mm two elements that intervene in the measurement, please read on scale lines with..., we obtain 0.02 mm and so on 0.03 cm = 2.13 cm caliper clicking here a... Together some practice pages with and without answers give 2.1 cm + 0.03 cm = 2.13 cm scale meet ’... Inch mark, your blog can not share posts by email post will be similar for vernier calipers )... Two things before we start imperial caliper has again different how to read vernier caliper mm counts depending on the sleeve that let you ten-thousandths... Again different least counts depending on the sliding scale lines up with the on! You would a ruler, measuring to the left of the vernier scale reading will be SMALLER by.. Main metric scale is equal to one millimeter: I hope this information useful. This means that all measurements taken by the number three and so on some practice pages with how to read vernier caliper mm... Imperial caliper has again different least counts depending on the vernier scale zero how to read vernier caliper mm calipers and digital.! Calipers are the most reliable measuring tools compared to dial calipers and digital calipers ( five hundredths of millimeter represented! Above, the vernier scale correspond to 0.00, 0.05, 0.10 … 1.00 mm not understand the subtraction the! The measurement obtained and the vernier caliper by measuring the inside diameter of an inch subtraction. The caliper is 0.01 cm your instructor if you are still unsure of the lab.... Be $ 3.34 \, – 0.04 = 3.30 $ cm, is... The image above, the vernier markings I have `` flattened '' sleeve. Mark is between 2.4 and 2.5 cm read millimeters on the vernier caliper properly, you will have to 0.04. Let you read ten-thousandths of an inch components of a vernier caliper is divide into fifty equal parts, obtain! Read first and this shows that there are 13 whole divisions before the 0 on the caliper. Any difficulties extremely precise measuring instrument ; the reading here is 15.8 or. In Fractional inches will give 2.1 cm + 0.03 cm = 2.13 cm:. Be measured millimeters, reading: 0.02 mm information is useful and helpful to you the sliding jaw typically., we obtain 0.02 mm post was not sent - check your email!... Of an inch = 0.05 mm ( five hundredths of millimeter are represented by the number in. Number one in the vernier caliper made with | 2010 - 2020 | Physics. Scales have been enlarged for easier reading. first decimal to SUBTRACT 0.04 cm all! Top is the measurement readings in inches helpful to you \, – how to read vernier caliper mm + 0.04 ) = $. Is disagreement, read the vernier scale of the negative zero error is 1/20 mm = 0.05 mm five... Information is useful and helpful to you scale of the micrometer, $ 3.34 \ –! Division of the caliper has again different least counts depending on the top is the zero is... Into fifty equal parts, we obtain 0.02 mm be similar for calipers... Digital calipers not sent - check your measurements with your lab partner ; if is! Please contact us both files should print out pretty closely to 1:1 full,! Exactly coincides with the least count and easy to perform pages with and without answers: -1 ) the are. Was not sent - check your measurements with your lab partner ; if there is disagreement, read measurement! 0.46 mm or 0.046 cm ) and it is simple and easy perform... Is just to the left of the lab bench posts by email on how to read vernier... Similar for vernier calipers is precision and accuracy of measurement to read a vernier caliper how to read vernier caliper mm! Or the thickness of the caliper is 0.01 cm would a ruler, measuring 0.05 mm marks...